Light source
halogen-xenon lamp
Reading	
4-channel (simultaneous reading on four optical channels)
Linear range
from -0.2 to 3.0A
Light filters	
340, 405, 450, 505, 545, 600, 630, 700 nm
Photometric accuracy	
± (1% of the reading +0.005A from 0 to 1.5A); ± (2% of the reading +0.005A from 1.5 to 3.0A)
Dosing volume	
25 μl – 1.95 ml
Number of microplates	
1 (96 wells)
Thermal control of plate	
25°C, 37°C or ambient temperature
Number of specimens	
96 (including calibrators and controls)
Number of reagents	
27 або 44 (if they are placed into a sample rack)
Washing	
automatic 8-probe
QC options	
storing control data, printing Levey-Jennings or QC range plots, calculating SDs
Software language	
English, Russian
Data output	
possibility of connection to the LIS (Laboratory Information System)
Power	
100 В-250 VAC/ 50-60 Hz
Operating conditions	
temperature: 5..40°C; humidity: less than 85%
Dimensions	
86 cm (W) x 51 cm (L) x 40 cm (H)
Weight	
≈ 35 kg