Productivity
300 tests / h (450 with ISE)
Methods of analysis
End point, kinetics, fixed time, 1-2 reagent, multistandard, bichromatic, blank reagent/serum, immunoturbidimetry
ISE
Optional
Number of sample positions
60 sample positions (including the standard, blank and control; serum position and initial test-tube are available)
Number of reagent positions
40 reagent positions (80 reagent positions are optional)
Sample volume
1-100 μl, step 0.1 μl
The volume of the reagent
R1: 1-400 μl, step 0.1 μl  R2: 1-400 μl, step 0.1 μl
Wavelength range
340 – 810 nm
Installed light filters
10 basic 340 nm, 405 nm, 450 nm, 505 nm, 546 nm, 578 nm, 620 nm, 670 nm, 750 nm, 800 nm and optional: 380 nm, 492 nm, 630 nm, 650 nm, 660 nm, 700 nm, 810 nm 810 nm
Dimensions, weight
850 mm x 720 mm x 530 mm; 90.0 kg