Throughput
60 tests per hour
Determined parameters 
22 blood parameters with 3-part WBC differentiation: WBC (total white blood cell count); LYM (lymphocytes count); MID (middle-sized cells): GRA (granulocytes count); LYM% (lymphocytes percentage); MID% (middle-sized cells percentage); GRA% (granulocytes percentage); RBC (red blood cell count); HCT (hematocrit); MCV (mean corpuscular volume); HGB (hemoglobin); MCH (mean corpuscular hemoglobin); MCHC (mean corpuscular hemoglobin concentration); PLT (platelet count); PCT (platelet percentage); MPV (mean platelet volume); P-LCC (macro platelet count); P-LCR (macro platelet percentage); RDW-CV (red cell distribution width); RDW-SD (red cell distribution width, standard deviation); PDW-CV (platelet distribution width); PDW-SD (platelet distribution width, standard deviation)
Sample volume
100 μl (whole blood, closed test tubes), 25 μl  (opened tubes) – optionally 
Sample input
Opened or closed tubes
Method of measurement
impedance method for measuring standard parameters; Photometric method for hemoglobin determination 
Calibration
automatic or manual calibration
Quality control parameters 
Up to 24 levels of QC available; mean, SD and CV for all parameters, Levy-Jennings curves ; QC data base
Measuring chambers
3 chambers for dilution of whole blood and counting + 1 needle cleaning
Apertures 
80 μl (RBC/PLT), 100 μl (WBC)
Coefficient of dilution 
Primary dilution - 1:160, total dilution RBC - 1:25 600, WBC dilution - 1:196
Memory, data storage
10 000 results, including histograms 3-diff WBC, RBC, PLT and QC data
Language of interface
English, Russian or others
Data input and output
Colorful touch-screen, Built-in thermal printer
Dimensions
31,6 x 49,2 x 36 cm; 17,8 kg