Performance
80 tests / hour
Measured parameters
22 parameters with leukocyte differentiation into 3 subpolations: WBC, LYM%, LYM, GRA, GRA%, MID, RBC, HGB, HCT, MCV, MCH, MCHC, RDWcv, RDWsd, PLT, MPV, PDWcv, PDWsd, PCT, P-LCC, P-LCC%, P-LCR
Sample volume
25 μl in normal mode and 50 μl in pre-dilution mode
Sample capture
open tube 
Measurement methods
volumetric impedance method for counting blood cells (WBC, RBC, PLT); photometric method for determining HGB
Calibration
automatic or manual
Quality control
separate database of quality control; Levi-Jennings charting
Dilution ratio
WBC, HGB 1 160 / RBC, PLT: 1: 32000
Measuring chambers
2 chambers for dilution of whole blood and counting
Apertures
70 μm (RBC / PLT), 100 μm (WBC / MIX)
Memory, data storage
10,000 results, including 3-diff histograms of WBC, RBC and PLT and quality control data
Interface language
English, others by request
Data input and output
color touch screen; built-in thermal printer
Sizes, weight
33 х 31 х 38 cm; 12 kg