Chemiluminescence Principle	
Direct chemiluminescence <br> Magnetic microbeads separation, fast and efficient <br> Key technology of ABEI-label and magnetic microbead separating <br> enhances stability and sensitivity of MAGLUMI reagents
Throughput
Throughput: up to 200 tests/hour <br> 24 hours ready to use
Reagent Loading
20 reagent positions on board with continuous loading <br> 24-hour refrigerated, constant 10°C±3°C in reagent area <br> RFID reading all information of reagent <br> Indicator light available, real-time status monitoring
Sample Loading
Up to 72 sample positions <br> Continuous loading, STAT available <br> Indicator light available, real-time status monitoring <br> Barcode recognition: Barcode reader available
Reagent Features
Calibrators and internal controls included <br> 10 points master curve, 2-points re-calibration <br> RFID tag storing all informations of reagents <br> Storage temperature 2°C-8°C
Reaction System
Single reaction cup with five-sided heating at 37.0°C (±0.3°C) incubation, ensuring accurate, quick and uniform incubation <br> Incubating 80 cuvettes simultaneously <br> Non-contact mixing with different modes before incubation, ensuring sufficient reaction
Sampling System	
Titanium needle for crushproof <br> Teflon coated for carry-over prevention <br> Liquid level detection, clot detection, prevent bubble interference, collision detection, wash automatically
Washer	
Intelligent 4-step washing technology magnetic field design ensure excellent separation <br> Resuspension washing mode efficiently reduces non-specific binding and improves sensitivity <br> 
Measuring Chamber	
Exclusive patented ABEI technology  <br> Heating and cooling bidirectional temperature control function included  <br> Independent measuring room, avoiding light cross-contamination
Software System	
Friendly software UI design <br> Real-time status display for each test procedure <br> Self-recovery function make sure the analyzer running smoothly without unnecessary manual intervention <br> Bidirectional LIS connection
Dimensions & Weight	
Dimensions: 78 × 90 × 75 сm <br> Weight: 127 kg