<h2 align="center">Model</h2>
<h2 align="center">BS3000T</h2>
Measurement parameters
Clinical chemistry   Coagulation  Specific proteins
Methods of analysis
Photometry: endpoint, fixed time, kinetics, blank, bichromatic, bireagent, multipoint, etc.  Coagulation: Nephelometry.   Specific proteins: nephelometry, kinetic nephlometry.
Wavelength range
Biochemistry: 340 nm, 405 nm, 492 nm, 510 nm, 546 nm, 620 nm, 670 nm + others on request.   Coagulation: 450 nm, 670 nm + others on request  Specific proteins: 450 nm, 670 nm + others on request.
Cuvettes
Flow cuvette, direct reading cuvette and a coagulation cuvette.
Working volume
50 – 500 μl
Thermostat
2 reagent positions; 15 positions of micro-test tubes; 12 positions of macro-tubes.
Quality control
QC of normal and pathological samples, automatic statistics and analysis of control data.
Memory
Storage of 1000 test results.
Display
7-inch color touchscreen
Test menu
Biochemistry: Renal function, liver function, cardiomarkers, minerals, blood disorders, lipidogram, DOA and others.   Coagulation: IF, ACTH, PM, fibrinogen, AT-III, D-dimer, and others.   Specific proteins: SRB, RF, ASLO, C3, C4, immunoglobulins, apolipoproteins, HbA1C, and others.
<h2 align="center">Model</h2>
<h2 align="center">BS3000C</h2>
Measurement parameters
Clinical chemistry
Methods of analysis
Photometry: endpoint, fixed time, kinetics, blank, bichromatic, bireagent, multipoint, etc.  Coagulation: Nephelometry.
Wavelength range
Biochemistry: 340 nm, 405 nm, 492 nm, 510 nm, 546 nm, 620 nm, 670 nm + others on request.   Coagulation: 450 nm, 670 nm + others on request. 
Cuvettes
Flow cuvette, direct reading cuvette.
Working volume
50 – 500 μl
Thermostat
2 reagent positions; 15 positions of micro-test tubes; 12 positions of macro-tubes.
Quality control
QC of normal and pathological samples, automatic statistics and analysis of control data.
Memory
Storage of 1000 test results.
Display
7-inch color touchscreen
Test menu
Biochemistry: Renal function, liver function, cardiomarkers, minerals, blood disorders, lipidogram, DOA and others.   Coagulation: IF, ACTH, PM, fibrinogen, AT-III, D-dimer, and others.
<h2 align="center">Model</h2>
<h2 align="center">BS3000B</h2>
Measurement parameters
Clinical chemistry   Specific proteins
Methods of analysis
Photometry: endpoint, fixed time, kinetics, blank, bichromatic, bireagent, multipoint, etc.   Specific proteins: nephelometry, kinetic nephlometry.
Wavelength range
Biochemistry: 340 nm, 405 nm, 492 nm, 510 nm, 546 nm, 620 nm, 670 nm + others on request.   Specific proteins: 450 nm, 670 nm + others on request.
Cuvettes
Flow cuvette, direct reading cuvette.
Working volume
100 – 500 μl
Thermostat
2 reagent positions; 15 positions of micro-test tubes; 12 positions of macro-tubes.
Quality control
QC of normal and pathological samples, automatic statistics and analysis of control data.
Memory
Storage of 1000 test results.
Display
7-inch color touchscreen
Test menu
Biochemistry: Renal function, liver function, cardiomarkers, minerals, blood disorders, lipidogram, DOA and others.   Specific proteins: SRB, RF, ASLO, C3, C4, immunoglobulins, apolipoproteins, HbA1C, and others.
<h2 align="center">Model</h2>
<h2 align="center">BS3000М</h2>
Methods of analysis
Endpoint, fixed time, multistandard, differential, bichromatic and immunoturbidimetric and others
Wavelength range
340, 405, 492, 510, 546, 578, 620 nm+ 2  additional filters on request.
Cuvettes
2 types: flow cuvette, direct reading cuvette
Working volume
200 – 800  μl
Incubator
20 incubation positions
Thermostatic control
Temperature: 25°С, 30°С,37°С.
Quality control
QC of normal and pathological samples, automatic statistics and analysis of control data.
Memory
More than 200 programs, 1000 results of analyzes
Display
Touch screen
Test menu
Biochemistry: Renal function, liver function, cardiomarkers, minerals, blood disorders, lipidogram, DOA and others.
<h2 align="center">Model</h2>
<h2 align="center">BS3000Р</h2>
Methods of analysis
Endpoint, fixed time, multistandard, differential, bichromatic and immunoturbidimetric and others
Wavelength range
340, 405, 492, 510, 546, 578, 620 nm+ 2  additional filters on request.
Cuvettes
Flow cuvette
Working volume
450 – 500  μl
Thermostatic control
Temperature: 25°С, 30°С,37°С.
Quality control
QC of normal and pathological samples, automatic statistics and analysis of control data.
Memory
More than 200 programs, 1000 results of analyzes
Display
LCD display
Test menu
Biochemistry: Renal function, liver function, cardiomarkers, minerals, blood disorders, lipidogram, DOA and others.